TECHWEIGH SCALE, LLC 909-433-0426 CALIBRATION
24670 MANTON ROAD FAX 909-433-0429 CERTIFICATE # 9999
COLTON, CA 92324
SYSTEM ID: 123456 TEST DATE: 11-11-11 RECALL DATE: 02-11-12
ASSET ID: 654321 SERIAL ID: 23456789-TES STATUS: ACTIVE
CUSTOMER: TESTER INCORPORATED BUILDING: 1
ATTENTION: JOSEPH TESTER III ROOM: 111 ADDRESS: 12345 W. ANYSTREET DEPARTMENT: PRODUCTION
CITY: ANYTOWN, CA CONTACT: JOE TESTER II
MFG: RICELAKE CAPACITY X DIVISION: 5,000 LB. X 1 LB
MODEL: IQ+355 TYPE: SCALE
APPLIED TOL: NIST 44 TABLE 6 CLASS III . CONDITION: GOOD
This test was conducted at 78 degrees Fahrenheit, 32% Relative Humidity, and 29.92 inHg.
TEST DATA Pre-Tol: WT Post Tol: WT Procedure: TWS 5.9
Applied Standard As Found Value As Adjusted Value Uncertainty Tolerance
25 LB. 25 LB. 25 LB. 0.139 G. 1 LB.
250 LB. 250 LB. 250 LB. 1.39 G. 1 LB.
500 LB. 500 LB. 500 LB. 2.78 G. 1 LB.
1000 LB. 1001 LB. 1000 LB. 5.56 G. 2 LB.
20,25,50 LB. CLASS F. CAST WEIGHTS STD MFG: RICELAKE
CAL DATE: 01/07/11 STD RECALL: 01/07/12 TEST NO: 8888/C NIST NO:111/123455-11
TEST TECH: JOHN SMITH APPROVED BY: JOE JONES
Signature line Signature line
Techweigh Scale, llc is ISO/IEC 17025, ANSI Z540, AND ISO 10012-1 Approved. This calibration is traceable to the national institute of standards (NIST) through documents on file at Techweigh Scale, LLC. This calibration only applies to the equipment that is listed on this certificate. This calibration certificate shall not be reproduced except in full, without the written approval of Techweigh Scale,llc.
QACC001 REV. G